Laser-induced breakdown spectroscopy (LIBS) uses a highly energetic laser pulse as an excitation source to ablate material from sample surfaces, vaporize them to plasma and then identify elements based on characteristic spectral emissions. This enables identification and removal of trace contaminants. In this case study, the LIBS technique was used for detection of low levels of silicone contaminants in CFRP composites.
Laser-induced breakdown spectroscopy (LIBS) uses a highly energetic laser pulse as an excitation source, to optically breakdown, vaporize and convert elements in the sample to plasma state. The amount of sample subjected to breakdown (termed ‘ablation’) and vaporization to plasma, is in the range of nanograms to picograms. The plasma cools, within microseconds and various atoms emit characteristic spectral lines that can identify them. The detector needs to be gated to capture full spectral information within the microseconds duration. The principle is illustrated in Figure 1.
Adhesive bonded fibre-polymer composites are important for applications such as light-weighting of structures and components in the transportation sector. They also find application in consumer goods, industrial and infrastructural applications. Composites can be bonded to each other or to metals or non-metals using adhesives. Adhesive bonds between surfaces require the surfaces to be free from contaminants. Even trace contaminants can reduce bond strength and result in structural failure under load.
Carbon Fibre Reinforced Polymers (CFRP) are an important category of composites, due to their light weight, strength and corrosion resistance. Due to its high sensitivity LIBS can be used to detect ultralow concentrations of contamination on CFRP samples.
In this case study, the LIBS technique was used for detection of low levels of silicone contaminants in CFRP composites. Sample (30.5 cm × 30.5 cm) CFRP panel surfaces were contaminated in a controlled manner with polydimethylsiloxane (PDMS). The panels were analyzed by LIBS before and after laser ablation, providing information on the ability of LIBS to remove silicone as well as to detect very low levels of silicone.
LIBS ablation was performed using an Nd:YVO4 laser system with pulse energies below 30 μJ, duration about 10 picoseconds, lasing wavelength of 350 nm and frequency about 400 kHz. For contamination studies, wavelengths ranging from 370 nm to 900 nm with a 10 nm step size were used at three incident angles: 65°, 70°, and 75°. From this data, the PDMS layer thickness was inferred. SEM micrographs of ablation depth of the surface at increasing energy of single pulse laser shots are seen in Figure 2. The craters produced show no melted or redeposited material around the crater border. Above15 μJ, an inner crater starts forming due to the increasing energy and the crater deepens faster than it widens. In contrast to the crater contour, the inner crater has a circular shape. In all the LIBS measurements, the pulses do not interact with the carbon fibres underneath the resin surface layer, as shown in the SEM micrographs.
Figure 1: The principle of LIBS and the type of spectrum for Si.
Figure 2: Single laser pulse shot, SEM micrographs of ablated craters at a) 7.5 μJ (3.77 J/cm2), b) 15 μJ (7.53 J/cm2), c) 25 μJ (12.55 J/cm2), and d) 30 μJ (15.06 J/cm2).
The figures 3 and 4 show peak heights for Carbon and Silicon, before and after ablation. The reduction in height of Si peak after ablation is obvious. The ability of LIBS to detect low levels of Si and to remove contaminant by ablation from CFRP is also established
Figure 3: LIBS spectra before laser ablation for the wavelength from 240 nm to 300 nm. The relevant peaks are 247.9 nm for C I, and 251.6 nm and 288.2 nm for Si I.
Figure 4: LIBS spectra after laser ablation for the wavelengths from 240 nm to 300 nm. The relevant peaks are 247.9 nm for C I, and 251.6 nm and 288.2 nm for Si I.
ASTM E572 test method covers the analysis of stainless and alloy steels by Wavelength Dispersive X-ray Fluorescence Spectrometry (WDXRF). It provides rapid, multi-element determinations with sufficient accuracy to assure product quality.
The ASTM D2674 is a standard test method for the analysis of sulfochromate etches solution used in the surface preparation of aluminum. The ASTM D2674 standard specifies a method for determining the efficacy of an etchant used to prepare the surface of aluminum alloys for subsequent adhesive bonding.
An immunological method for quantization of Hevea Natural Rubber (HNRL) proteins using rabbit anti-HNRL serum. Rabbits immunized with HNRL proteins react to the majority of the proteins present, and their sera have the capability to detect most if not all the proteins in HNRL.
ASTM G65 measures the resistance of metallic materials to abrasion using the dry sand/rubber wheel apparatus. The quality, durability, and toughness of the sample are determined using this test. Metallic materials are ranked in their resistance to scratching abrasion under a controlled environment.
ASTM E2141 test methods provide accelerated aging and monitoring of the performance of time-dependent electrochromic devices (ECD) integrated in insulating glass units (IGU). This test helps to understand the relative serviceability of electrochromic glazings applied on ECD.
ASTM C724 test method is used in analyzing the quality and ease of maintenance of a ceramic decoration on architectural-type glass. This test method is useful in the acknowledgment of technical standards.
Send us a request
Process for testing
You share material and testing requirements with us
We ensure your sample pick-up in an ensured manner
We deliver test report to your inbox
Just share your testing requirements and leave the rest on us!
Free, no-obligation consultation
Quick turnaround time
Let us combine our capabilities to achieve success!!