Raman spectroscopy is a versatile vibrational spectroscopic technique, used to identify and quantify molecular species, based on Raman Scattering. A promising application of Raman spectroscopy is for on-line monitoring of water pollution, including detecting complex dissolved contaminants. This case study illustrates the potential of Raman spectrometry for on-line analysis of water pollution, caused by dissolved pharmaceutical drugs, pesticides or salts coming from fertilisers.
Raman Spectroscopy is a vibrational spectroscopic technique, based on the Raman effect and is used for investigating molecular structures. When light is inelastically scattered by matter, the wavelengths of the scattered photons are different from the incident photons. The wavelength shift (or frequency shift) is related to the molecular vibrations, which are unique to each molecular structure. The change in frequency of scattered photons (termed Raman shift), may be positive or negative. Raman spectroscopy involves detecting and recording these wavelength shifts and using them to analyse what is in the sample. A Raman spectrum will have several peaks, each peak corresponding to a specific mode of molecular vibration of a chemical species. The location of the maximum peak leads to species identification, while its intensity value correlates with species concentration. While the basic technique has evolved significantly and there are a myriad of applications, one high potential area of application of conventional Raman spectroscopy is in water pollution monitoring.
With the synthetic chemical industry constantly introducing new molecules that eventually find their way into surface waters as effluents, the effective surveillance of water pollution has become challenging. The traditional grab sampling and batch analysis from end of pipe or final sink, is not suitable, especially where trace contaminants are involved. It is more effective to have real-time, on-line analysis methods. This case study illustrates the potential of Raman spectrometry for on-line analysis of water pollution, caused by dissolved pharmaceutical drugs, pesticides or salts coming from fertilisers.
This case study considers examples representing three categories of important organic pollutants, namely drugs, pesticides and chemical fertilisers, for in-situ analysis by Raman Spectroscopy. The study on drug related molecules was performed on ethinylestradiol (a hormone used in contraceptives and a drug based on niflumic acid (used in anti-inflammatory agents). Among pesticides glyphosates are common and hence considered for this study. With regard to fertilisers, Nitrates are a recognized source of eutrophication of waterways due to agricultural run-off, hence a nitrate was considered for analysis.
Measurements on the polluted water samples were made with a BWTek iRamanTM spectrometer. The excitation wavelength was 532 nm, with a 4 cm-1 spectral resolution. The spectral range covered was 175–4000 cm-1. Calibration was done and then polluted samples analysed. Using this set-up, in this illustrative case study, the limit for Nitrate detection was 100 mg/l while Glyophosate was detectable down to 10 mg/l. The limit achieved for niflumic acid needs improvement, as the minimal concentration this experimental set-up and method could detect was 2500 mg/l.
Figure 1 shows the Raman spectra of a contraceptive pill whose main component is ethinylestradiol and also of a drug based on niflumic acid. For clarity, only the peaks of the main molecules are marked. Figure 2 shows the Raman spectra of a typical dissolved synthetic fertilizer mixture comprising 150 g/l of Potassium Phosphate and 89 g/l of Sodium Nitrate.
Figure 1: Normalized Raman Spectra of a drug containing Ethinylestradiol and another containing Niflumic acid, dissolved in demineralized ASTM D1193 type IV reagent water.
Figure 2: Normalized Raman Spectra of a synthetic salt mixture comprising 150 g/l of Potassium Phosphate and 89 g/l of Sodium Nitrate dissolved in demineralized ASTM D1193 type IV reagent water.
Ivana Durickovic, Mario Marchetti, Raman spectroscopy as a polyvalent alternative for water pollution detection, IET Science, Measurement and Technology, 2014, Vol. 8, Issue 3, pp. 122–128.
ASTM E572 test method covers the analysis of stainless and alloy steels by Wavelength Dispersive X-ray Fluorescence Spectrometry (WDXRF). It provides rapid, multi-element determinations with sufficient accuracy to assure product quality.
The ASTM D2674 test is a standard test method for the analysis of sulfochromate etch solutions used in the surface preparation of aluminum. The ASTM D2674 standard specifies a method for determining the efficacy of an etchant used to prepare the surface of aluminum alloys for subsequent adhesive bonding.
An immunological method for quantization of Hevea Natural Rubber (HNRL) proteins using rabbit anti-HNRL serum. Rabbits immunized with HNRL proteins react to the majority of the proteins present, and their sera have the capability to detect most if not all the proteins in HNRL.
ASTM G65 measures the resistance of metallic materials to abrasion using the dry sand/rubber wheel apparatus. The quality, durability, and toughness of the sample are determined using this test. Metallic materials are ranked in their resistance to scratching abrasion under a controlled environment.
ASTM E2141 test methods provide accelerated aging and monitoring of the performance of time-dependent electrochromic devices (ECD) integrated in insulating glass units (IGU). This test helps to understand the relative serviceability of electrochromic glazings applied on ECD.
ASTM C724 test method is used in analyzing the quality and ease of maintenance of a ceramic decoration on architectural-type glass. This test method is useful in the acknowledgment of technical standards.
Send us a request
Process for testing
You share material and testing requirements with us
We ensure your sample pick-up in an ensured manner
We deliver test report to your inbox
Just share your testing requirements and leave the rest on us!
Free, no-obligation consultation
Quick turnaround time
Let us combine our capabilities to achieve success!!