ASTM E110-14 Test for Rockwell and Brinell Hardness of Metallic Materials by Portable Hardness Testers
ASTM E110-14 test method is used to determine Metals' Rockwell and Brinell Hardness by using Portable Hardness Testers. Portable hardness testers test metals that cannot be tested by fixed instruments because of metal's size, location, or other requirements.
ASTM E110-14 method defines the requirements for portable instruments that measure metals’ Rockwell or Brinell hardness by indentation tests outside of a lab when specimens cannot be tested by a standard E10 or E18 hardness tester due to their size or weight. Hardness testing at a part of the specimen does not represent the physical characteristics of the entire specimen.
ASTM E10 and E18 are the standard tests for measuring Brinell and Rockwell hardness, respectively, but sometimes, because the specimen is too big and heavy, it cannot be tested with their fixed instruments. ASTM E110-14, instead, uses a portable hardness tester. But the principles used behind this test are the same as E10 and E18.
ASTM E18 Rockwell Hardness Test Principle:
First, An indenter applies a minor load on the metal, causing a dent. The depth of the dent establishes the baseline depth. Without removing the minor load, a major load is applied for a specific time, and the depth of the indentation produced by it is measured. The harder the metal, the shallower the penetration, and vice versa. The Rockwell hardness value is derived from the difference in the two depths: the baseline depth and final depth. This distance is converted to a hardness number.
ASTM E10 Brinell Hardness Test Principle:
Brinell hardness is determined by forcing a hardened steel or carbide ball of known diameter under a known load into a surface and measuring the diameter of the indentation with a microscope.The diameter is used to derive the Brinell hardness value. The portable hardness tester should be verified before a hardness test with the appropriate standard, E10 or E18. This is done to ensure that environmental conditions (temperature), damage during travel and usage have not affected the ability of the test equipment to perform properly.
There is no standard shape or size for a Brinell test specimen, but the test specimen should conform to the following: The specimen should be thick enough that no bulge appears on the part of the specimen opposite the indentation when the load is applied. The thickness of the specimen should be at least ten times the depth of the indentation. The minimum width should conform to the requirements for indentation spacing.
1. Rockwell Hardness:
HR = N – hd
Where, d is the depth in mm (from the zero load point), and N and h are scale factors that depend on the scale of the test being used.
2.Brinell hardness number:
BHN = Brinell hardness number
P = applied load
D = diameter of the indenter ball in mm, and
d = measured mean diameter of the indentation in mm
ASTM E572 test method covers the analysis of stainless and alloy steels by Wavelength Dispersive X-ray Fluorescence Spectrometry (WDXRF). It provides rapid, multi-element determinations with sufficient accuracy to assure product quality.
The ASTM D2674 is a standard test method for the analysis of sulfochromate etches solution used in the surface preparation of aluminum. The ASTM D2674 standard specifies a method for determining the efficacy of an etchant used to prepare the surface of aluminum alloys for subsequent adhesive bonding.
An immunological method for quantization of Hevea Natural Rubber (HNRL) proteins using rabbit anti-HNRL serum. Rabbits immunized with HNRL proteins react to the majority of the proteins present, and their sera have the capability to detect most if not all the proteins in HNRL.
ASTM G65 measures the resistance of metallic materials to abrasion using the dry sand/rubber wheel apparatus. The quality, durability, and toughness of the sample are determined using this test. Metallic materials are ranked in their resistance to scratching abrasion under a controlled environment.
ASTM E2141 test methods provide accelerated aging and monitoring of the performance of time-dependent electrochromic devices (ECD) integrated in insulating glass units (IGU). This test helps to understand the relative serviceability of electrochromic glazings applied on ECD.
ASTM C724 test method is used in analyzing the quality and ease of maintenance of a ceramic decoration on architectural-type glass. This test method is useful in the acknowledgment of technical standards.
Send us a request
Process for testing
You share material and testing requirements with us
We ensure your sample pick-up in an ensured manner
We deliver test report to your inbox
Just share your testing requirements and leave the rest on us!