Coulometric and Volumetric Determination of Moisture in Plastics ASTM D6869-17
The ASTM D6869-17 test method is used for coulometric and volumetric detection of moisture in polymers using the Karl Fischer Reaction (the reaction of iodine with water). Moisture content does have an impact on the physical qualities of various polymers.
This ASTM D6869-17 test technique is designed to determine moisture content in most polymers. To quantify the amount of moisture in a polymer sample, this approach employs the interaction of iodine (I2) with water (Karl Fischer Reaction). Plastics having volatile components, such as residual monomers and plasticizers, release substances that interfere with the I2/water reaction.
As per ASTM D6869-17, the processability of various plastics will be influenced by moisture. Surface flaws (such as splay or bubbling) or hydrolysis deterioration are caused by high moisture content. Polymerization is caused by a combination of low moisture and high heat. This method can be used to measure moisture in the range of 0.005% to 100%.
For ASTM D6869-17, the samples are heated to release water, which is then carried to the titration cell by a nitrogen carrier gas. The reaction of water with I2 is used to determine the moisture collected in the solution within the titration cell. Instrumented approaches are used to detect endpoints. The interaction of I2 with water is used to determine the amount of moisture present. With 10.71 Coulombs of producing current equating to 1 mg of water, coulometric instruments employ Faraday’s equation to quantify the moisture present. Whereas, volumetric devices measure the volume of I2 solution necessary to maintain a constant current.
For ASTM D6869-17, specimens in a variety of forms are allowed, including moulded powder, moulded shapes, and re-grind. Prior to testing, it is recommended that moulded specimens be chopped into smaller portions (recommended maximum size 4 by 4 by 3 mm).
As per ASTM D6869-17, automatic calculations are performed by commercial coulometric devices depending on the micrograms of water measured. The sample type, sample weight, and % moisture are reported.
ASTM E572 test method covers the analysis of stainless and alloy steels by Wavelength Dispersive X-ray Fluorescence Spectrometry (WDXRF). It provides rapid, multi-element determinations with sufficient accuracy to assure product quality.
The ASTM D2674 is a standard test method for the analysis of sulfochromate etches solution used in the surface preparation of aluminum. The ASTM D2674 standard specifies a method for determining the efficacy of an etchant used to prepare the surface of aluminum alloys for subsequent adhesive bonding.
An immunological method for quantization of Hevea Natural Rubber (HNRL) proteins using rabbit anti-HNRL serum. Rabbits immunized with HNRL proteins react to the majority of the proteins present, and their sera have the capability to detect most if not all the proteins in HNRL.
ASTM G65 measures the resistance of metallic materials to abrasion using the dry sand/rubber wheel apparatus. The quality, durability, and toughness of the sample are determined using this test. Metallic materials are ranked in their resistance to scratching abrasion under a controlled environment.
ASTM E2141 test methods provide accelerated aging and monitoring of the performance of time-dependent electrochromic devices (ECD) integrated in insulating glass units (IGU). This test helps to understand the relative serviceability of electrochromic glazings applied on ECD.
ASTM C724 test method is used in analyzing the quality and ease of maintenance of a ceramic decoration on architectural-type glass. This test method is useful in the acknowledgment of technical standards.
Send us a request
Process for testing
You share material and testing requirements with us
We ensure your sample pick-up in an ensured manner
We deliver test report to your inbox
Just share your testing requirements and leave the rest on us!