In-Situ tribological testing of Acrylonitrile Butadiene Rubber (NBR) elastomers under realistic High pressure gas environments is demonstrated in this case study. The equipment used is a Linear Reciprocating Tribometer housed in an autoclave. Hydrogen effects on NBR are apparent from increased friction and wear, compared to high pressure Argon or ambient air.
The tribological behaviour of contacting elastomer and metal surfaces in high pressure gas applications depends upon the type of gas and pressure temperature envelopes. Dynamic loads on the materials, due to pressure and thermal cycling need consideration. Viscoelastic behaviour of elastomers affects the coefficient of friction and elastomer wear-rate increases by adhesion to the metal surface. Hydrogen induced degradation of elastomeric materials is a concern for elastomers deployed in Hydrogen infrastructure. Tribological testing in high pressure Hydrogen environments, simulating actual operating conditions, provides realistic performance data for elastomeric components in high pressure Hydrogen service.
In-Situ tribometers enclose the test samples, tribometry equipment and sensors within a pressure vessel, mitigating safety risks due to high pressure hydrogen. In-Situ Linear Reciprocating tribometry enables high pressure tribological tests of linearly sliding elastomer- metal surfaces in a Hydrogen environment.
Acrylonitrile Butadiene Rubber (NBR) is used for many dynamic sealing applications in high pressure Hydrogen dispensing systems. Typical applications include NBR seals for Hydrogen compressors, seals and stem packings in isolation and control valves, diaphragms and actuators. NBR elastomers are known to degrade in Hydrogen environments due to gas permeation and dissolution. In dynamic sealing applications of NBR, wear and tear present additional concerns.
This case study uses a bespoke pin on flat type linear reciprocating tribometer system to investigate the tribological performance of NBR in high pressure Hydrogen, Argon and Air environments. Controlled frictional loading is provided by a ball at the tip of a steel pin pressing upon an NBR sample moving in a direction perpendicular to the loading force, at various normal loads, velocities and frequencies. In this case, a disk-shaped sample of NBR is clamped horizontally and the weight loaded ball runs linearly, back and forth, across the sample surface to form a wear track. Figure 1 (A) illustrates the principle, while Figures (B) and (C) are a photograph of the apparatus.
NBR Samples of 1/8th inches (3.175 mm) thickness with 40A Shore hardness rating were cleaned and subjected to tribological tests in ambient air at atmospheric pressure, Argon at 24 MPa and Hydrogen at 28 MPa. Wear measurements from the in-situ linear variable differential transformer (LVDT) were confirmed by ex-situ measurements done subsequently, using optical profilometry. It was observed that sample wear tracks in the high-pressure Hydrogen tests showed more pitting, depressions, and cracking, than those from High pressure Argon testing. Additionally, the coefficient of friction was maximum for the Hydrogen exposed samples.
Enter Sample and testing requirementsProvide your contact information
EELS analysis of gate and channel is performed on fin field-effect transistors (finFETs). Scanning transmission electron…
FTIR analysis is used to study the migration and leaching of phthalate plasticizers from p-PVCs. Phthalate…
Nano-scale surface roughness is a critical parameter in fabricated thin-films that are used in optics, solar…
ASTM E572 test method covers the analysis of stainless and alloy steels by Wavelength Dispersive X-ray Fluorescence Spectrometry (WDXRF). It provides rapid, multi-element determinations with sufficient accuracy to assure product quality.
The ASTM D2674 test is a standard test method for the analysis of sulfochromate etch solutions used in the surface preparation of aluminum. The ASTM D2674 standard specifies a method for determining the efficacy of an etchant used to prepare the surface of aluminum alloys for subsequent adhesive bonding.
An immunological method for quantization of Hevea Natural Rubber (HNRL) proteins using rabbit anti-HNRL serum. Rabbits immunized with HNRL proteins react to the majority of the proteins present, and their sera have the capability to detect most if not all the proteins in HNRL.
ASTM G65 measures the resistance of metallic materials to abrasion using the dry sand/rubber wheel apparatus. The quality, durability, and toughness of the sample are determined using this test. Metallic materials are ranked in their resistance to scratching abrasion under a controlled environment.
ASTM E2141 test methods provide accelerated aging and monitoring of the performance of time-dependent electrochromic devices (ECD) integrated in insulating glass units (IGU). This test helps to understand the relative serviceability of electrochromic glazings applied on ECD.
ASTM C724 test method is used in analyzing the quality and ease of maintenance of a ceramic decoration on architectural-type glass. This test method is useful in the acknowledgment of technical standards.
You share material and testing requirements with us
You ship your sample to us or arrange for us to pick it up.
We deliver the test report to your email.
Let’s work together!
Share your testing requirements with us and we will be happy to assist you.
What Material or product do you have?
What analysis do you need?
How many parts or coupons do you have?
How fast do you need the results back?
Do you know the goal of the analysis you need?