ASTM E1787-16 Standard test method for Anions in Caustic Soda and Caustic Potash by Ion Chromatography
Anionic impurities in caustic soda and caustic potash are monitored by manufacturers and users for quality control of the products. Anions of primary interest are chloride, chlorate, and sulfate. The ASTM E1878-16 test method can determine precision estimates for these three impurities.
ASTM E1787-16 is used to determine anionic impurities in 50% Caustic Soda (Sodium hydroxide) and 50 % Caustic potash (Potassium hydroxide) solutions using ion chromatography (IC). Anions that can be determined at a concentration of 0.1 to 100 ppm include bromide, chlorate, chloride, fluoride, nitrate, phosphate, and sulfate. By varying sample size we can use this method for anhydrous caustic soda and caustic potash.
In ASTM E1787-16 initially the ion chromatograph and neutralization device is set as per the manufacturer’s instructions. The detector range is changed as per concentration of anions in the sample. The system is equilibrated by pumping eluent through the columns and the detector until a stable baseline is obtained (approximately 15 to 20 min). Sample solutions are prepared by taking the known weight of samples in a 100 mL volumetric flask and diluted to volume using water. The dilution ratio depends upon the anions concentrations in the sample. The neutralization module is capable of neutralizing a solution containing 25% sodium hydroxide when the solution is cycled through the neutralizer at least twice. Sample solutions containing up to 50 g of 50 % product or 25 g of anhydrous sample, therefore, can be neutralized effectively. The sample is injected into an ion chromatograph and the ion chromatogram of anions is recorded. Figure 1 shows an IC chromatogram of 45 % caustic potash that was diluted 20 g/100mL.
Figure 1: Typical Chromatogram of 45 % Caustic Potash (Diluted 20 g/100 mL) Using AS-12A Separator Column and 2.7 mmol/L Na2CO3/ 0.3 mmol/L NaHCO3 Eluent.
50 % concentration solution of Caustic soda and Caustic potash.
ASTM E1787-16 calculates the concentration of each anion of interest using the peak area of that anion from the sample chromatogram and response factor derived from “best-fit” linear equation obtained during the calibration procedure.
A = area of anion peak from the chromatogram of the diluted sample, area counts,
B = final volume of diluted sample, mL,
R = response factor from “best-fit” calibration line, (μg/mL)/area counts, and
G = sample weight, g.
Concentration of the sodium or potassium salt of the anion can be calculated as follows:
Hence, we can determine the anions concentration in caustic soda and caustic potash using ion Chromatography using ASTM E1787-16 method.
ASTM E572 test method covers the analysis of stainless and alloy steels by Wavelength Dispersive X-ray Fluorescence Spectrometry (WDXRF). It provides rapid, multi-element determinations with sufficient accuracy to assure product quality.
The ASTM D2674 test is a standard test method for the analysis of sulfochromate etch solutions used in the surface preparation of aluminum. The ASTM D2674 standard specifies a method for determining the efficacy of an etchant used to prepare the surface of aluminum alloys for subsequent adhesive bonding.
An immunological method for quantization of Hevea Natural Rubber (HNRL) proteins using rabbit anti-HNRL serum. Rabbits immunized with HNRL proteins react to the majority of the proteins present, and their sera have the capability to detect most if not all the proteins in HNRL.
ASTM G65 measures the resistance of metallic materials to abrasion using the dry sand/rubber wheel apparatus. The quality, durability, and toughness of the sample are determined using this test. Metallic materials are ranked in their resistance to scratching abrasion under a controlled environment.
ASTM E2141 test methods provide accelerated aging and monitoring of the performance of time-dependent electrochromic devices (ECD) integrated in insulating glass units (IGU). This test helps to understand the relative serviceability of electrochromic glazings applied on ECD.
ASTM C724 test method is used in analyzing the quality and ease of maintenance of a ceramic decoration on architectural-type glass. This test method is useful in the acknowledgment of technical standards.
Send us a request
Process for testing
You share material and testing requirements with us
We ensure your sample pick-up in an ensured manner
We deliver test report to your inbox
Just share your testing requirements and leave the rest on us!
Free, no-obligation consultation
Quick turnaround time
Let us combine our capabilities to achieve success!!